Every HeartStart HS1 comes ready-to-rescue. The device is pre-installed in the carrying case and battery and pads are pre-installed in the device. Simply pull up the handle on the front of the AED to activate it.
Guided audio instructions
Talks you through your training
The HeartStart HS1 provides on-demand audio coaching. Calm, clear voice instructions are tailored to the responder’s actions, providing AED and CPR guidance during the resuscitation of a sudden cardiac arrest victim.
Smart Biphasic therapy
Safely delivers maximum shock strength
Philips SMART Biphasic therapy uses a unique combination of high current - to maximize effectiveness, and a lower energy dose - to minimize side effects that are harmful to a fragile heart.
Built-in self test
Ready when needed
Your HeartStart HS1 goes through a 120-point test before it leaves the factory. On the job, it conducts self tests every single day, not just weekly. It performs over 80 different tests in all. Even the pads are tested for readiness.
Quick Shock feature
Minimize chest compression interruptions
CPR is even more vital to survival than previously realized. Rapidly delivering a shock after chest compressions is critical. The HeartStart HS1’s Quick Shock feature reduces the time between hands-off and shock delivery.
Integrated SMART pads
Senses application to patient skin The pads are "smart" because they can sense that they have been removed from the cartridge and applied to the patient. This helps the HeartStart OnSite know at what point the user is in the response process.
Clear, intuitive icon driven operation
The HeartStart HS1’s graphics-based visual instructions are designed to communicate instantly, with minimum reading and interpretation. Users not fluent in the device’s language can follow easily.
Defibrillator, battery (one, preinstalled), SMART Pads (one, preinstalled), Set-up/Maintenance guide with expiration date tags, Owner’s Manual, Quick Reference Guide.
HeartStart Onsite (HS1) Ready-Pack option R01: Defibrillator inside carry case, battery (one, pre-installed), SMART Pads (one set pre-installed plus spare set in carry case), Set-up/Maintenance guide with expiration date tags, Owner’s Manual, Quick Reference Guide.
Truncated Exponential Biphasic. Waveform parameters adjusted as a function of each patient’s impedance.
Adult defibrillation peak current: 32A (150J nominal) into a 50 ohm load. Pediatric defibrillation (with optional Infant/Child SMART Pads Cartridge): 19A (50J nominal) into a 50 ohm load.
Shock-to-Shock Cycle Time
Typically less than 20 seconds between shocks in a series.
Able to deliver a shock after the end of a CPR interval, typically in eight seconds.
Detailed voice messages guide responder through use of the defibrillator.
Instructions for adult and infant/child CPR available at user’s option. Shock Delivery Via adhesive pads placed on patient’s bare skin as illustrated on pads.
Green SMART Pads cartridge handle, green On/Off button, blue i-button, orange Shock button.
Ready light; blue i-button; caution light.
Solid objects per EN60529 class IP2X Drip-proof per EN60529 class IPX1
Operating: 32º - 122º F (0º - 50º C)
Standby: 50º - 109º F (10º - 43º C)
Operating: 0% to 95% relative, non-condensing
Standby: 0% to 75% relative, non-condensing
Operating: 0 to 15,000 ft (0 to 4,500 m)
Standby: 0 to 8,500 ft > 48 hours and 8,500 to 15,000 feet < 48 hours
Abuse Withstands 1 meter drop to any edge, corner or surface.
Meets EN1789 random and swept sine, road ambulance specification in operating and standby states.
Meets EN55011 Group 1 Level B Class B and EN61000-4-3.
Patient Analysis System
Evaluates patient ECG to determine if a rhythm is shockable. Rhythms considered shockable are ventricular fibrillation (VF) and certain ventricular tachycardias (VT) associated with lack of circulation. For safety reasons, some VT rhythms associated with circulation will not be interpreted as shockable, and some very low-amplitude or low-frequency rhythms will not be interpreted as shockable VF.
Meets AAMI DF80 guidelines and AHA recommendations for adult defibrillation (Circulation 1997;95:1677-1682).
The effects of pacemaker artifact and electrical noise are minimized with artifact detection.
9 Volt DC, 4.2 Ah, composed of disposable long-life lithium manganese dioxide primary cells. Capacity Minimum 200 shocks or 4 hours of operating time (EN 60601-2-4:2003)
Battery is labeled with an install-by date of at least five years from date of manufacture.
Four years typical when battery is installed by the install-by date. (Will power the AED in standby state within the specified standby Temperature range, assuming one battery insertion test and no defibrillation uses.)
Adult SMART Pads Cartridge
M5071A defibrillation pads for patients 8 years of age and older or 55 lbs. (25 kg) and over.
Infant/Child SMART Pads Cartridge
M5072A defibrillation pads for patients under 8 years of age or 55 lbs. (25 kg). Rx only.
Adult: nominal 150 Joules into a 50 ohm load
Infant/Child: nominal 50 Joules into a 50 ohm load
Disposable cartridge, containing adhesive defibrillation pads, clicks into defibrillator for an integrated pads solution.
Active Surface Area
13.2 in² (85 cm²) each
Adult pads: 54 in (137.1 cm)
Infant/Child pads: 40 in (101.6 cm)
Cartridge is labeled with a use-by date of at least two years from date of manufacture.
Adult Training Pads Cartridge
Infant/Child Training Pads Cartridge
Special pads put HeartStart OnSite into training mode and disable its energy delivery capability. Training pads feature 8 real-world training scripts. Used with training mat (included) or with adapters on manikins.
Specifically tests readiness-for-use of pads (gel moisture).
Battery Insertion Test
Upon battery insertion, extensive automatic self-tests and user-interactive test check device readiness. Status Indicator Blinking green "Ready" light indicates ready for use. Audible “chirp” indicates need for maintenance.
Data Recording and Transmission
Wireless transmission of event data to a PC or Palm® PDA, using the IrDA protocol.
First 15 minutes of ECG and the entire incident’s events and analysis decisions.